Can you tell me what size inlet fitting is provided with the SDAFMPANEL? My customer needs to know what type of hose fitting to use to connect to the flow meter. Would you also have a drawing showing how the SDAFMPANEL connects to the ADM Aspida?
The tubing should be a 6mmOD x4mmID PVC tubing.
Inlet is the bottom spigot on the rear of the flow meter panel, outlet is the top spigot on the rear of the flow meter panel.
Connection to the ADM Aspida should be made from the top spigot on the rear of the flow meter panel, then route through the hole to the left of the display and connect to the flow adaptor which then plugs into the front of the ADM Aspida.
The ADM Aspida can then be mounted to the panel as described in the attached Quick Start Guide.