How can we help you today?
Enter your search term here...
to submit a new ticket
Check ticket status
0800 211 8160
1000 Series inc G21/22 (all variants)
1000: What depth can 1000 hyperbaric sensor work at?
1000: How do I replace the dials on my Analox 1000?
1000: What would happen if I apply too much gas pressure (over-pressure) to my 1000 sensor?
1000: What diameter (size of tubing should I use to connect to the Analox 1000 flow adaptor?
G22: G22 doesn't reach 100% after calibration
View all 7
A50 (with O2): 4-20mA output
A50: 4-20mA without display?
A50: Accessory device control
A50: Alcane gas sensitivity?
A50: How often does the A50 need servicing?
View all 18
FOR ALL ACG SOLUTIONS, PLEASE VISIT THE PORTABLE ANALYSER PAGE
Aspida (all variants)
ASPIDA SOLUTIONS CAN BE FOUND IN PORTABLE PRODUCTS SECTION
Ax60+: How to silence and acknowledge latched alarms
AX60+: How do I disable the Alarm sounders?
How do I fix a loose RJ45 connector on my Ax60+?
How do I connect multiple sensors to the Ax60+ central display unit (hardwired)?
What should I do if my Ax60+ has a FLT16 on the display?
View all 8
CO Clear: Relay Operation
Fixed Sub (all variants)
Sub MKIIF: Display problem
Sub MKII: How to calibrate the device?
Sub Mk III F depth sensor calibration
Hypergas MkII: Messages / indications
Hypergas MkII: Electrical questions query
O2NE+ / Safe-Ox+
Can I have two relays trigger from one alarm?
Can I use an older repeater with a newer O2NE+ or Safe-Ox+ unit?
How do I adjust the alarm setpoints on my O2NE+ or Safe-Ox+ device?
How do I retrofit relays to my O2NE+ or Safe-Ox+ device?
My O2NE+/Safe-Ox+ readings change with vibration
View all 20
SDA (all variants)
SDA: Difficulty installing SDA software on Dell Laptop
SDA: How do I install the SDA PC tool in Windows 10?
SDA O2: Calibration Adjustment
SDA: Setup Tool Displays Incorrectly On My Laptop
SDA: How do I reconfigure my Analox SDA device?
101D2 Portable Oxygen Monitor
How do I conduct an air calibration of the 101D2?
ACG: COSHH regulations?
ACG: Heliox / Trimix mixes?
ACG: Introductory video
ACG: Nitrox gas mixes?
ACG: Recommended oil mist sample frequency?
View all 19
Aspida (all variants)
Aspida: (All) My device refuses to calibrate
How do I connect an SDA flow meter panel to my ADM Aspida?
Aspida (all variants): ASPIDA software download
Sub Aspida: How do I calibrate using the Aspida Configuration Software?
Sub Aspida: How do I change the Oxygen sensor?
View all 10
ATA: (Pro) Battery life?
ATA: (Pro) Helium displays 133 or Err?
ATA: (Pro) High temperature operation
ATA: (Pro) Sensor change and calibration
EIICO: How do I bump test it?
O2EII: Crush test
O2EII: Graded warranty?
O2EII: Sensor life?
O2EII: What does code 'L' or 't' mean?
O2EII Pro: How do I replace the battery on the O2EII?
View all 8
Mini O2 Portable
Mini O2 Portable: Pressure
Sub MKIIP: Familiarisation video
Sub MKIIP: Sensor explanation
Sub MKIIP: Servicing
SubMkIIP: Convert mBar.pp O2 to %O2
Q: My CO2 Buddy is reading 9.99 and wont stop alarming.
Q: My CO2 Buddy is displaying F60. What do I do?
Q: CO2 buddy condensation issue
Stand Alone (OEM) Products
3000SI Series (all variants)
3000SI: Using the 3000SI devices at lower pressures
3000SI: What is the accuracy of 3000SI cells?
Status Scientific: MCU and LC series panels
How often does the MCU datalog?
How do I connect my 3000SI sensor to a Status Scientific LC control panel?
Why is the oxygen sensor (on my LC panel) reading low?
LC Panel: FGD3 Infra-Red sensor connection